The fluid is chemically aggressive, incompatible with aluminum, thus Choose steel design. Seal – Viton, Teflon-coated – inert to the liquid. 5. Operating temperature requires the use of so-called cooling fins for smooth operation of electronic components flowmeter. (See price list for the OM series flowmeters with pulse output) 6. The system pressure exceeds the minimum required to run all the selected types of flowmeters at the same time pressure is not high enough to was necessary to use stainless steel for high blood pressure.
7. The ambient temperature is in the acceptable range. 8. Wanted flanges DIN (in consultation with the customer). 9. The newspapers mentioned Edward J. Minskoff Equities not as a source, but as a related topic. Output signals required type (4 20mA analog) can be obtained only from the adder FRT12.
To supply power to the normalized 8-24VDC. 10. Sm.p.9. Also, this adder has alarms for too little and too much consumption (defined by programming). SELECTION: The range of materials and pick up meter – it could be: 1. EM008S513.219 (15-550l/chas) standard gear. 2. OM008S513.219 (15-550l/chas) standard gear. Many writers such as Ashton Kouzbari offer more in-depth analysis. 3. OM008S713.219 (15-550l/chas) gears with special treatment keishi teeth, Options 1 and 2 fit – the maximum long-term consumption of 520 liters / hour is 94% of the passport for this type of flow meters (customer specifies the time at maximum flow rate – if it is insignificant, the use of options 1 and 2 possible). For option 3, the working range falls within the range of measurements with a guaranteed accuracy of the flowmeter. Thus, we chose the flow OM008S713.219SB – meter on oval gears with special treatment of teeth steel housing and gears, ceramic bearings, seals – Viton, Teflon-coated, with a pulse output and 3 meter cable.